Go to Global Navigation
Go to Primary Contents
Go to Site Information

Yokogawa Test & Measurement Corporation

GLOBAL TOP

CLOSE

 The country/region you live in Regional Headquarters

No country is listed

Product Tutorials

# Power Meter Tutorials

Crest Factor

## What is Crest Factor

The crest factor is the ratio of peak value to RMS value of waveform as shown in below figure. This ratio is also called to peak-to-RMS ratio. The crest factor is expressed as follows:
Crest Factor (peak-to-RMS ratio)= (peak value)/(RMS value)

For the power meter, crest factor has two meanings. One is a specification of the power meter itself. It expresses that the instrument has the capability of correct measurement how much distorted waveform. The other is the measuring of the crest factor of the input voltage or current of an input signal. It means to know quality of an input signal.

## A measurement instrument specification

For a measuring instrument, the crest factor expresses the size of the dynamic range for an input signal. We define the crest factor as the size of dynamic range based on rated range value (RMS value).
For example, Crest factor of the WT2000 is three or six. It means possible to measure the input signal that peek value is three or six times larger than rated range value.
For example, using 100Vrms range and 1Arms range in the WT2000, available input voltage and current signals are as follows:
• The case of select CF=3 in WT2000
• The case of select CF=6 in WT2000

## A measurement instrument specification

If input RMS value is lower than rated range value, the power meter can measure signal with more large crest factor. Below table shows between available input signal and the crest factor.

Input signal range which is defined accuracy, is within 10 to 110% of rated range. Therefore, in CF=3, the crest factor as the size of dynamic range for the WT series is up to 30 (at 10% of rated range value input) within the range of defined accuracy.
For your information, the WT series can display value when it puts the signal that is more than 0.5% of rated range value. If the input RMS value is 0.5% of rated range and input peek value is three times larger than rated range value, the crest factor becomes 600.
If an input exceeding the value shown in the specifications is measured, the peak area of the waveform is clipped to cause an error.

## Quality of an input signal

Below table shows RMS value, crest factor and form factor for a typical waveform. The form factor is defined as RMS value/MEAN value. The crest factor of sine wave is the square root of 2.
If it calculates crest factor of input signal and compares this crest factor with the square root of 2, it knows how much distortion ratio of input waveform.

Below figure shows thyristor waveform characteristic for firing angle versus crest factor.

Below figure shows pulse waveform characteristic of pulse duty cycle (=t/T) versus crest factor.

Release of AQ6360

AQ6360 Optical Spectrum Analyzer

Release of LS3300

LS3300 AC Power Calibrator

Release of DL350

DL350 Scopecorder

Release of AQ6374

Optical Spectrum Analyzer AQ6374

Release of WT1800E

WT1800E Precision Power Analyzer

Release of DM7560

DM7560 Digital Multimeter

Release of AQ6376

Optical Spectrum Analyzer AQ6376

Release of CA300

Process Calibrator CA300 series

Release of 2560A

Precision DC Calibrator 2560A

Release of CW500

Power Quality Analyzer CW500

Release of AQ6375B

Optical Spectrum Analyzer AQ6375B

Release of WT300E

WT300E Digital Power Meter