FUKUHARA Tatsuya1 TSUCHIYA Tsuyoshi1 WATANABE Junichi1 SUZUKI Shinji1
We have developed DXAdvanced DX1000/DX2000 which has substantially strengthened the basic performance and features of the DAQSTATION DX series. DXAdvanced has realized multiple channels and increased measurement interval speed, with DX1000 having up to 12 measurement channels and DX2000 having up to 48 measurement channels. If connecting to an external DAQMASTER MX100 Series unit or the like, DX2000 has a maximum measurement capacity of 348 CH with 1 sec being the shortest measurement interval. In addition to the reinforcements made to the basic performance with display variations, network features, increased memory capacity and so on, the new DXAdvanced is housed in a drop-proof structure that protects it from dust.
This paper outlines the strengthened measurement capabilities, the display variations, the increased memory capacity, the network and so on, and introduces the PC software.
- Industrial Automation Product Business Center
|Figure 1 External View of
the DX1000 and DX2000
Yokogawa Electric Corporation, one of the leading manufacturers of industrial recorders, put the DAQSTATION DX series stations on the market in 1999, which added information processing and transmission functions to conventional recorders. Since then, Yokogawa has developed the CX series incorporating control functions, and the DX-P series incorporating enhanced security functions and conforming to 21CFR Part 11 in order to respond to diversifying field needs. These products have been very well received in the market.
This time around, Yokogawa has developed the DXAdvanced series DX1000 and DX2000, which replace the conventional DX100/200 series as successor models and provide substantially enhanced measurement, display, memory, and networking functions and performance in order to respond to market needs. Figure 1 shows their external view of the DX1000 and DX2000. The DXAdvanced series is designed as a next-generation data acquisition station, which incorporates advanced networking functions, a wide variety of display functions, and sophisticated information processing functions.
- Measurement; Multichannel Support and Increased Speed
Figure 2 Channel Extension
Using MW100 Series Units
In addition, connecting the DX2000 series to external input units such as DAQMASTER MW100 series units enables analog measurements on up to 348 channels (with the shortest interval of 1 second) to be made (Figure 2).
- Display and Operability
UBS keyboards and infrared remote control units are supported, which makes text input and configuration quick and easy. A wide variety of display screens have been inherited and operability has been improved by adding unique functions such as a function to customize the operation menu that tends to become complicated due to sophisticated functions, and a function to jump to a preset operating display screen by just touching the button.
Compact flash cards up to 200 MB (approximately 170 times the conventional memory) have been made possible to be mounted as the internal memory for long-time recording of measured data on multiple channels. In addition to a highly reliable compact flash card (1 GB), USB memory is supported as external media for enabling a large amount of data to be recorded and offline data to be transferred to a PC.
Figure 3 Drawn-out Internal
Chassis of the DX1000N
The dust- and drip-proof structure of the front panel designed in compliance with the IEC529-IP65 standard and the sealed structure of the casing on the backside of the unit allow for the use under harsh environments. The log-in function and the key-lock function installed in the unit prevent unauthorized persons from accessing the unit, and the locking mechanism installed for the front door latch, which requires a special key, prevents any person from mistakenly touching the power switch or accessing the external media.
In addition to the above, the use of a temperature- compensated clock IC achieved a clock accuracy of ± 10 ppm (approximately one-tenth of the conventional accuracy). As a sister model of the DX1000, the DX1000N is available. Its internal chassis is designed to be drawn out of the casing to facilitate maintenance such as periodic calibration after wiring, or regular maintenance (Figure 3).
Table 1: Number of Measurement Channels and Measurement Intervals of DX1000/DX2000
|Model||Maximum number of measurement channels||Shortest measurement interval (ms)|
In the DXAdvanced station, the functions became sophisticated from those in the existing series, and the manufacturing cost was reduced by reviewing each of the electrical and electronic circuits compared with the existing models. Figure 4 shows the hardware configuration of the DXAdvanced station.
Figure 4 Block Diagram of the DXAdvanced Station
- Input Measuring Section
Figure 5 Signal Conditioner Chip (SCC)
- CPU System Section
The CPU system section incorporates a high speed 32-bit RISC CPU (180-MHz operating frequency). When the power is turned on, the CPU program within the Flash ROM isloaded to DRAM on the CPU and started by the boot program. The CPU is connected to each dedicated control IC for USB and Ethernet communications, to SRAM that serves as the primary buffer for configuration information and measured data, and to internal and external memory for saving data (CF-CARD), via the bus interface ASIC. The input measuring section (SCC) and serial signals from the real time clock (RTC IC) used for the purpose of clock are also connected via the ASIC in the same manner. The use of a temperature-compensated crystal oscillator in the real time clock (RTC) has improved the absolute time accuracy for recording data (Figure 6).
- Power Supply Section
The power supply section that used to use a power supply purchased from a third party has been newly designed in- house. As a result, the degree of design freedom for heat dissipation in the sealed enclosure was increased and the service life of the component having a recommended replacement interval (electrolytic capacitor) was extended, and components needed to suppress an increase of heat were simplified. Thus, it became possible to use the identical power supply section both in the DX1000 and DX2000.
- Display Section
The DX1000 uses a 5.5-inch and the DX2000 a 10.4-inch wide view TFT color LCD display respectively.
Figure 6 Clock Accuracy vs Ambient Temperature
A new component layout and architectural redesign for the sealed enclosure reduced the influence on the reference junction compensation error, which is a problem in multipoint temperature measurements using thermocouples. In addition, the use of an aluminum plate in conjunction with a heat conductive sheet disperses heat and suppresses heat build-up in the CPU, which is expected to occur due to an increase in the CPU utilization as a result of sophistication of functions.
- Conformance to Overseas Standards
The series comply with a number of overseas standards such as CE marks, CSA, UL (NRTL/C) and acquired the relevant certifications. Furthermore, the series pays attention to environmental design and employs gold plating in printed circuit boards in preparation for global environmental laws and rules, which may have to be addressed in the future.
- Structure of Internal Chassis Drawing-out for Model DX1000N
A panel- mounted instrument is usually fixed using the clamping fixtures after it is inserted through the panel hole from the front. Then, wires are connected to the terminals on the backside of the instrument before it is used. Therefore, whenever regular calibration is carried out, wires must be disconnected on the backside of the instrument before the instrument is taken out for calibration. To simplify regular calibration, the internal chassis drawing-out model DX1000N has been developed based on the DX1000. The DX1000N is designed such that the internal chassis, which comprises an A/D circuit section, power supply section, CPU circuit section and display section, all relating to calibration, can be taken out of the front panel without disconnecting terminal wiring on the backside of the instrument.
- Memory Management Function
A compact flash (CF) card was adopted as the internal memory to allow for recording data on more channels for a longer time. The CF card has error check and correction (ECC) and access equalization processing functions for the purpose of increasing reliability.
The CF card is internally divided into two segments, for saving data and saving files depending on the usage. The data saving segment is used to save data such as measured data that are currently being sampled, as well as alarm and message data. Fixed size files are available for each type of data to increase efficiency in save-processing. The data saving segment employs SRAM as the primary buffer (Figure 4) to buffer a certain amount of data before writing it to the segment, in order to reduce the write frequency and increase the durability of the CF card, which has a limited service life of writing.
The file saving segment is used to save various types of data held in the data saving segment as data files that are created over a certain period of time. Data can be output to the outside without data conversion because the data are saved in the internal memory as files. Furthermore, the file saving segment uses a ring buffer to process files, so a particular file can be loaded to the external storage media any number of times, unless the file is overwritten.
The CF card used as the internal memory is controlled by the File Allocation Table (FAT) filing system. A FAT segment is available in SRAM (Figure 4) and a journaling function was installed to prevent the file system from breaking down in a power failure. Thus, after power recovers, data can be restored using the FAT data and operating history data that were recorded before the power failure occurred, and the consistency of the file system can be maintained.
- Japanese Input and Chinese Input Functions
Provision of a Kanji (Chinese characters) input function allows for inputting Kanji and Hiragana (JIS first level only) for messages and tag names. Kanji input is accomplished by converting a clause string of input single-byte kana characters and selecting appropriate kanji characters.
In addition, a Chinese input function is available for the Chinese market. The function converts tone 'pinyin'- characters into written Chinese language. The Chinese input function was developed by a development team in the People's Republic of China, which was representative for all the development work from the specification of input system to design, coding, and evaluation.
PC SOFTWARE - DAQSTANDARD FOR DXAdvanced
DAQSTANDARD for DXAdvanced is PC software that comes standard with the DXAdvanced series, and that provides a function that allows users to easily configure the settings.
- Maintaining Compatibility
The configuration software also supports the settings of the existing DX100/DX200 series. In addition, Operability of the display screens of the DX100 and DX200 series have been inherited.
- Arithmetic Expression Input Function
Arithmetic expression input can be accomplished by just selecting function names and channels using the mouse.
This paper outlines the distinguished functions of the DXAdvanced DX1000 and DX2000, and hardware, firmware and PC software that are needed for those functions. We are convinced that the DX1000 and DX2000 with substantially enhanced functions will be able to play a central role in the market of small- scale instrumentation systems. We would like to pay attention to changes in the market and continue developing products that can respond to a wide range of user needs in the future.
- TAKAHASHI Masahiko, KATSUOKA Shuji, HOSAKA Yuji, KAKIHARA Kaname, "DAQSTATION DX100/ DX200 Series of Paperless Recorders", Yokogawa Technical Report, Vol.44, No. 1, 2000, pp. 37-40 in Japanese
- MORI Sadao, FURUYA Takahito, HIRAI Kazuo, "Hardware of PC-based High Speed Data Acquisition Unit DAQMASTER MX100", Yokogawa Technical Report, Vol.49, No. 1, 2005, pp. 27-30 in Japanese
- DAQSTATION', 'DXAdvanced', and 'DAQMASTER' are registered trademarks of Yokogawa Electric Corporation. In addition, product names, software names and other names that appear in this document are trademarks or registered trademarks of their respective company or organized body.
In the mid 1970s, Yokogawa entered the power business with the release of the EBS Electric Control System. Since then, Yokogawa has steadfastly continued with the development of our technologies and capabilities for providing the best services and solutions to our customers worldwide.
Yokogawa has operated the global power solutions network to play a more active role in the dynamic global power market. This has allowed closer teamwork within Yokogawa, bringing together our global resources and industry know-how. Yokogawa's power industry experts work together to bring each customer the solution that best suits their sophisticated requirements.
Water & Wastewater
Water resources are finite, and therefore contributing to a sustainable water cycle is one of the Sustainable Development Goals (SDGs). Yokogawa has been providing advanced digital control solutions for the stable supply of clean and safe water, wastewater treatment for protecting the water environment, water loss management and optimization of plant operation for reducing CO2 emissions and running costs. With our leading-edge technologies, dependable products and extensive expertise and experience of diverse water projects around the world, we work with you to provide sustainable water solutions that boost your business and add value throughout the plant lifecycle.
Yokogawa supports a wide range of water control applications in both the municipal and industrial water markets.
Zugehörige Produkte & Lösungen
DX1000/DX2000 mit Tastenbedienung
Die Daqstation-DX1000/2000-Serie ist ein Datenerfassungssystem mit Anzeige. Sein HMI unterstützt benutzerdefinierte Grafiken für Industrieanwendungen, Audit Trails und erweiterte Sicherheit nach FDA CFR21 Part 11.
Digitalschreiber sind voll integrierte registriertechnische Stationen mit Bildschirmanzeige, die mit einer sicheren, internen Datenspeicherung und Netzwerkanbindung punkten. Die Digitalschreiber sind allesamt NEMA-konform und binden die Datenerfassungssysteme in ein Steuerpult ein.