I want to connect the DX2000 and MW100 via Ethernet to acquire data from the MW100 onto the DX2000. Does the MW100 require the MATH function? Also, if I connect a power monitor to the DX2000, this is RS-422 communication, but I want to connect to the DX2000 via Modbus/RTU, so can I use both side-by-side?
The DX2000 (client) requires the expansion channel function option (suffix code: /MC1) or the MATH function option (suffix code: /M1). The MW100 (server) does not require the MATH function. Both serial and Ethernet can be used together. In that case, the DX2000 will require the RS-422A/485 communication interface option (suffix code: /C3) and the MATH function option (suffix code: /M1).
Súvisiace produkty a riešenia
Button Operated DX1000/DX2000
The Daqstation DX1000/2000 series is a data acquisition system with display. Its HMI supports custom graphics for industry applications, audit trail and advanced security conforming to FDA 21CFR Part 11.
For industrial DAQ applications, the MW100 offers scalability and can operate either standalone or integrated as a node within a larger automated system including SCADA or DCS.