The PH450 pH analyzer provides the best accuracy in the industry, featuring an advanced temperature compensation functionality, preloaded calibration standards, stability checks, and online sensor and analyzer diagnostics to provide verifiable results.
This four-wire, multivariable analyzer combines pH with temperature and ORP (Redox) measurement, all of which can be utilized through the different output functions: two mA current outputs, four independent SPDT contact outputs, and HART. Dynamic sensor checking and wash cycle functions assure trouble free and accurate analysis with minimum maintenance.
The PH450 has a high-resolution, simple touchscreen menu structure with five language options, so information is clearly visible and easily accessible to the operator. Simply select the language of choice and on-screen instructions assure that the best configuration for the application is obtained.
The PH450 offers self-diagnostics and process temperature compensation to provide accurate pH measurements in high purity water. One of these temperature compensation methods, the NEN6411 algorithm, takes into account the dissociation of water in strong acid and alkaline solutions, which is especially useful for pH measurement in Pure and Ultrapure water as found in Power Generation Water Chemistry.
|Input ranges||pH: -2 to 16 pH
ORP: -1500 to 1500 mV
rH: 0 to 100 rH
Temperature: -30 to 140 deg C (Pt1000)
|Transmission signals||General: Two isolated outputs of 4 to 20 mA DC with common negative. Maximum load 600 Ω. Bi-directional HART® digital communication, superimposed on mA1 (4 to 20 mA) signal.
Output function: Linear or Non-linear (21-step table) output for pH, temperature, ORP or rH.
Control function: PID control.
Burn out function: Burn up (21.0 mA) or burn down (3.6 mA) to signal failure accorded with NAMUR NE43.
Hold: The mA-outputs are frozen to the last/fixed value during calibration/commissioning.
|Contact outputs||General: Four SPDT relay contacts with display indicators.
Maximum values 100 VA, 250 V AC, 5 A
Maximum values 50 W, 250 V DC, 5 A
Status: High/Low process alarms, selected from pH, ORP, rH and temperature. Configurable delay time and hysteresis. Failure annunciation.
Control function: On/Off, PID duty cycle or pulsed frequency control.
Wash: Contact can be used to start manual- or interval time wash cycles.
Hold: Contact can be used to signal the Hold situation.
Fail: Contact S4 is programmed as fail-safe contact.
|Contact input||Remote wash cycle start.|
|Temperature compensation Function||Automatic or manual. Compensation to Nernst equation. Process compensation by configurable temperature coefficient, NEN6411 for water or strong acids/bases or programmable matrix.|
|Calibration||Semi-automatic 1 or 2 point calibration using pre-configured NIST, US, DIN buffer tables 4, 7 & 9, or with user defined buffer tables, with automatic stability check. Manual adjustment to grab sample.|
|Display||Graphical Quarter VGA (320 x 240 pixels) LCD with LED backlight and touch screen.|
|Housing||Cast Aluminum housing with chemically resistant coating; Polycarbonate cover with Polycarbonate flexible window, Protection IP66/NEMA 4X/CSA Type 3S|
Ratings; 100 to 240 V AC, Acceptable range; 90 to 264 V AC
Ratings; 50/60 Hz, Acceptable range; 50 Hz ±5%, 60 Hz ±5 %
Power Consumption; 15 VA
Ratings; 12 to 24 V DC, Acceptable range; 10.8 to 26.4 V DC
Power Consumption; 10 W
|Environment and operational conditions||Ambient temperature: -20 to 55℃ (-5 to 130℉)
Storage temperature: -30 to 70℃ (-20 to 160℉)
Humidity: 10 to 90 % RH at 40℃ (100℉) (non-condensing)
|pH input||≤0.01 pH|
|ORP input||≤1 mV|
|Temperature||≤0.3℃ (≤0.4℃ for Pt100)|
|Step response||< 4 sec for 90% (pH 7 - pH 4)|
Between measuring plant and control rooms, especially when the distance between these places is greater than the length of the standard appropriate electrode cables, the connecting equipment can be an expedient method for connecting sensor cables to a measuring instrument.
Wet scrubbers are used in utilities, paper mills, and chemical plants to remove sulfur dioxide (SO2) and other pollutants from gas streams. Undesirable pollutants are removed by contacting the gases with an aqueous solution or slurry containing a sorbent. The most common sorbents are lime, Ca(OH)2, and limestone, CaCO3.
For control of batch neutralization, a pH measurement coupled with a timer-controlled chemical feed scheme provides very satisfactory results.
This system can be adapted for either acid waste or alkaline waste neutralization.
The term "cooling tower" is used to describe both direct (open circuit) and indirect (closed circuit) heat rejection equipment. Cooling towers are heat-transfer units, used to remove heat from any water-cooled system. The cooled water is then re-circulated (and thus, recycled) back into the system. Since the process water is re-circulated, the mineral concentration increases as a result of the evaporation.
Industry:Refining, Food and Beverage, Power, Oil and Gas, Pulp and Paper, Chemical
The proliferation of microorganisms and the resultant formation of slime is a problem which commonly occurs in aqueous systems. Problematic slime producing microbes may include bacteria, fungi and/or algae. Slime deposits typically occur in many industrial aqueous systems including cooling water systems, pulp and paper mill systems, petroleum operations, clay and pigment slurries, recreational water systems, air washer systems, decorative fountains, food, beverage, and industrial process pasteurizers, sweetwater systems, gas scrubber systems, latex systems, industrial lubricants, cutting fluids, etc.
Industry:Refining, Food and beverage, Power, Oil and Gas, Pulp and Paper, Chemical
There are a number of suppliers of oil and fat products used for edible purposes. These products include, but are not limited to olive oil, peanut oil, soybean oil, sunflower oil, lard, shortening, butter, and margarine. The raw materials for these products include animal by-products, fleshy fruits (palm and olive), and oilseeds.
Industry:Food and Beverage
One of the primary applications for high purity water is for boiler feed water. The measurement of pure water pH can be one of the quickest indicators of process contamination in the production or distribution of pure water. Effective chemical treatment of the feed water is vital in maintaining the useful operating life and minimizing maintenance costs of the boiler.
Industry:Power, Pharmaceutical, Common
Process liquid analyzers such as pH meters, conductivity meters, ORP meters, and density meters play an important role at electrolysis plants in the control of concentrations of various process solutions. This requires both precision and stability under harsh conditions that include highly corrosive substances, high temperatures, and many impurities.
Most zinc are produced at hydrometallurgically, where a high-grade zinc product can be obtained and valuable metals mixed in the raw material can be recovered. In the hydrometallurgy, the raw material of zinc concentrate is roasted and then dissolved in sulfuric acid to remove impurities. The process called leaching and pH control of the leachate is important.
Problems at the wet end of a paper machine can rarely be corrected down stream. That is why monitoring and controlling pH in pulp stock is critical to the paper making process. Essentially, at every stage in the manufacture of paper, correct pH values play a vital role. Variations in the pH value at the head box have a negative effect on the quality of the paper produced.
Cyanide-bearing wastewater from mining and electroplating facilities and certain types of chemical plants is toxic and must be treated by oxidation with chlorine or chloride to bring the cyanide concentration within regulatory limits.
Industry:Electrical and Electronics
Fish perform all their bodily functions in water. Because fish are totally dependent upon water to breathe, feed and grow, excrete wastes, maintain a salt balance, and reproduce, understanding the physical and chemical qualities of water is critical to successful aquaculture. To a great extent water determines the success or failure of an aquaculture operation.
The treatment of wastewater from pulp and paper plants is a serious environmental concern. Yokogawa's submersion holder with an ultrasonic+air-jet cleaner (customized product) can reduce the manual cleaning frequency to just once every one or two months.
Industry:Pulp & Paper
Continuous technology improvement is ongoing in the pulp & paper industry to obtain the best possible performance. Problems at the wet end (stock preparation) can rarely be corrected downstream. That is why monitoring and controlling pH in pulp stock is critical to the paper making process. Essentially, at every stage in the manufacture of paper, correct pH values play a vital role.
Reverse osmosis (RO) is a separation process that uses pressure to force a solution through a membrane that retains the solute on one side and allows the pure solvent to pass to the other side. More formally, it is the process of forcing a solvent from a region of high solute concentration through a membrane to a region of low solute concentration by applying a pressure in excess of the osmotic pressure.
Continuous technology improvement is ongoing in the pulp & paper industry to obtain the best possible performance. The improved plant performance translates to the higher quality improvement and lower cost, and simultaneously environmental friendly plant operation.
Wastewater from electroplating facilities and certain types of chemical plants contains toxic forms of hexavalent chromium such as chromate and dichromate. The hexavalent chromium in this wastewater must be reduced before the water can be discharged. This requires a two-step process: hexavalent chromium (CR6) is reduced to trivalent chromium (CR3); and CR3 is precipitated as chromium hydroxide.
Industry:Electrical and Electronics
Removal of free oil and grease from a wastewater stream reduces the potential for equipment problems to occur further downstream. There are three forms of oil encountered in wastewater treatment at a refinery.
Sour Water is the wastewater that is produced from atmospheric and vacuum crude columns at refineries. Hydrogen sulfide and ammonia are typical components in sour water that need to be removed before the water can be reused elsewhere in the plant. Removal of these components is done by sending the sour water from the process to a stripping tower where heat, in the form of steam, is applied.
Power plant boiler houses designed to burn coal or high sulfur oil are required by Federal and State pollution regulations to "scrub" (remove) sulfur dioxide from flue gasses to meet emission limits. SO2 in flue gasses is known to be harmful to the environment, as it is one contributor to the formation of acid rain. pH control is critical for the proper functioning of the scrubber system.
Sodium chlorate is an inorganic compound with the chemical formula NaClO3. It is a white crystalline powder that is readily soluble in water. It is hygroscopic. It decomposes above 300 °C to release oxygen and leave sodium chloride. Several hundred million tons are produced annually, mainly for applications in bleaching paper.
In flue gas desulfurization systems that use magnesium hydroxide (Mg(OH)2) slurry, the consumption of the desulfurization agent (Mg(OH)2) is controlled by using online pH analyzers. A great concern in the pH measurement is heavy staining of the pH electrodes by the Mg(OH)2 slurry. To ensure accurate measurement, frequent cleaning of the electrodes with an acid is required, adding to both maintenance workload and cost.
The lifetime of a pH sensor has a significant impact on the overall annual costs of a pH measuring loop. Optimizing four key factors will decrease these costs and optimize process control and overall plant efficiency.
Looking for more information on our people, technology and solutions?