In the input range settings, you can add a specific value in the same manner as a bias setting by selecting a scale.
Ex) Set a thermocouple Type-K span of 0-100 DEGC. To display the temperature 0.5DEGC higher, set the following in the input range settings:
Mode = Scale; Type = TC; Range = K, Span lower limit = -0.5; Span upper limit = 99.5; Scale lower limit = 0.0; Scale upper limit = 100.0.
Gerelateerde producten & oplossingen
Button Operated DX1000/DX2000
The Daqstation DX1000/2000 series is a data acquisition system with display. Its HMI supports custom graphics for industry applications, audit trail and advanced security conforming to FDA 21CFR Part 11.
Removable Chassis DX1000N
The DX1000N features a removable chassis via the front panel of the instrument. This allows easy access and maintenance of internal components of the DX1000N.