In Basic setting mode, I set the Tag No. Use/Not setting to Use, but I don't set the tag number string (I enter all spaces), and this is supposed to be the setup to display channel numbers and tag comments (referring to section 5.2, “Displaying Tags or Channel Numbers” of the user's manual (IM 04L41B01-01E)). However, only the channel numbers are showing up on the actual screen.
With that setting, both the channel number and tag comments may not be able to fit in the limited display area, and only the channel number is displayed. To display the tag comments, set Tag No. Use/Not to “Not.”
Gerelateerde producten & oplossingen
Button Operated DX1000/DX2000
The Daqstation DX1000/2000 series is a data acquisition system with display. Its HMI supports custom graphics for industry applications, audit trail and advanced security conforming to FDA 21CFR Part 11.
Removable Chassis DX1000N
The DX1000N features a removable chassis via the front panel of the instrument. This allows easy access and maintenance of internal components of the DX1000N.