Managing Sterilization through Calculation of the Fo Value

### Overview

 Sterilization heating temperature measurement

In the sterilization process during pharmaceutical and food manufacturing, the "Fo" value is used as a benchmark for the sterilization time. The Fo value expresses the heating time required for disinfection, and varies according to fluctuations in heat applied to the substance being sterilized. The DX automatically calculates this Fo value from the heating temperature (the temperature measured on the DX), and is useful in managing the sterilization process by displaying and recording the value together with the heating temperature. Such careful monitoring and control allows minimization of sterilization time while maintaining the nutritional properties of the food.

Customer Needs

• To automatically calculate the Fo value, record it along with heating conditions, and use these as QC data
• To monitor the Fo value by heating temperature, shorten the heating process, and stabilize food quality
• To install a simple, low cost, highly reliable system without using external instruments or computers

### Process Outline

 Displays and records the heating temperature and Fo value

The heating temperature of the substance being sterilized is measured on the DX Paperless Recorder. The Fo value is automatically calculated according to the heating temperature (temperature measured on the DX) by the DX's mathematical functions. Computed results are displayed and recorded together with heating temperature and other parameters, and saved as quality data.

Also, the DX/AS1 supports 21CFR Part 11, and allows you to save formal pharmaceutical sterilization management data for approvals.

Definition of the Fo Value
Fo= Δt ∑ 10(T-Tb)/Z

Δt Measurement interval Heating temperature (temperature measured on the DX) 121.1°C (for steam pasteurization) Temperature unit of logarithmic sterilization capability changes (generally, 10°C is used).

Example of Calculation of the Fo Value on the DX
31= (K01*(K03**((01-K02)/K03)))/K04+31

01 Heating temperature (temperature measured on the DX) Fo value calculation result 1 sec or 0.125 sec (DX's measurement interval) 121.1°C (Fixed. Equivalent to Tb in the Fo equation) 10°C (Fixed. Equivalent to Z in the Fo equation) 60 (sec, for conversion to units of minutes)

### Yokogawa's Solution

 DX

DX series paperless recorders capable of automatically calculating the Fo value.

• Fo value calculated automatically by a math function (optional)
• Measured and calculated results saved electronically to local media while being displayed in real time on the large, color TFT display
• Also enables data sharing on the network (Ethernet compatible)

### Conclusion

DX series instruments allow you to automatically calculate and process Fo values as manufacturing/quality data at a reasonable cost. Also, by constantly ascertaining the Fo value according to the heating temperature, you can optimize (shorten) the heating process, and in the case of foodstuffs, minimize excessive heating which can damage the nutritional elements in foods.

• ### Voedsel en dranken

In de voedings- en drankenindustrie worden steeds hogere eisen gesteld aan de kwaliteit, veiligheid en beveiliging. Fabrikanten streven voortdurend naar kostenverlaging om de concurrentiepositie te verbeteren. De industriële IoT-oplossingen (IIoT) van Yokogawa maken dagelijkse verbeteringscycli mogelijk, elimineren defecte producten en minimaliseren de productiestilstand.

Meer

• ### Button Operated DX1000/DX2000

The Daqstation DX1000/2000 series is a data acquisition system with display. Its HMI supports custom graphics for industry applications, audit trail and advanced security conforming to FDA 21CFR Part 11.

Meer
• ### Removable Chassis DX1000N

The DX1000N features a removable chassis via the front panel of the instrument. This allows easy access and maintenance of internal components of the DX1000N.

Meer
• ### Panel Mount Recorders

Panel mount recorders are fully integrated data acquisition and display stations with secure, built-in data storage and network connectivity. Panel-mount solutions are NEMA compliant and integrate the data acquisition equipment into a control panel.

Meer
×