Managing Sterilization through Calculation of the Fo Value

### Overview

 Sterilization heating temperature measurement

In the sterilization process during pharmaceutical and food manufacturing, the "Fo" value is used as a benchmark for the sterilization time. The Fo value expresses the heating time required for disinfection, and varies according to fluctuations in heat applied to the substance being sterilized. The DX automatically calculates this Fo value from the heating temperature (the temperature measured on the DX), and is useful in managing the sterilization process by displaying and recording the value together with the heating temperature. Such careful monitoring and control allows minimization of sterilization time while maintaining the nutritional properties of the food.

Customer Needs

• To automatically calculate the Fo value, record it along with heating conditions, and use these as QC data
• To monitor the Fo value by heating temperature, shorten the heating process, and stabilize food quality
• To install a simple, low cost, highly reliable system without using external instruments or computers

### Process Outline

 Displays and records the heating temperature and Fo value

The heating temperature of the substance being sterilized is measured on the DX Paperless Recorder. The Fo value is automatically calculated according to the heating temperature (temperature measured on the DX) by the DX's mathematical functions. Computed results are displayed and recorded together with heating temperature and other parameters, and saved as quality data.

Also, the DX/AS1 supports 21CFR Part 11, and allows you to save formal pharmaceutical sterilization management data for approvals.

Definition of the Fo Value
Fo= Δt ∑ 10(T-Tb)/Z

Δt Measurement interval Heating temperature (temperature measured on the DX) 121.1°C (for steam pasteurization) Temperature unit of logarithmic sterilization capability changes (generally, 10°C is used).

Example of Calculation of the Fo Value on the DX
31= (K01*(K03**((01-K02)/K03)))/K04+31

01 Heating temperature (temperature measured on the DX) Fo value calculation result 1 sec or 0.125 sec (DX's measurement interval) 121.1°C (Fixed. Equivalent to Tb in the Fo equation) 10°C (Fixed. Equivalent to Z in the Fo equation) 60 (sec, for conversion to units of minutes)

### Yokogawa's Solution

 DX

DX series paperless recorders capable of automatically calculating the Fo value.

• Fo value calculated automatically by a math function (optional)
• Measured and calculated results saved electronically to local media while being displayed in real time on the large, color TFT display
• Also enables data sharing on the network (Ethernet compatible)

### Conclusion

DX series instruments allow you to automatically calculate and process Fo values as manufacturing/quality data at a reasonable cost. Also, by constantly ascertaining the Fo value according to the heating temperature, you can optimize (shorten) the heating process, and in the case of foodstuffs, minimize excessive heating which can damage the nutritional elements in foods.

• ### 制药

制药公司必须更好地使用和监控制造数据，实时改善质量管理，从而提高灵活性和敏捷性。制药专家与横河电机合作，一起研发生产解决方案，提供安全可靠的药品。我们共同借助于数字化转型和制造进步，满足监管要求，确保质量，加快上市时间，从而稳定、可靠地为患者供应药品。

更多
• ### 食品和饮料

在横河电机，我们了解当今的食品和饮料公司在气候变化、消费者需求和全球竞争力增强方面面临着前所未有的挑战。克服这些挑战需要创新的解决方案，需要重点关注生产、资产管理以及食品安全和质量等关键领域。

更多

• ### 底盘可拆卸型DX1000N

DX1000N特点是仪器的前面板有一个可拆卸的底盘，这使得DX1000N的内部组件易于连结和维护。

更多
• ### 按钮操作式DX1000/DX2000

Daqstation DX1000/2000系列是带显示的数据采集系统。HMI支持工业应用、审计跟踪和符合FDA 21CFR Part11高级安全性的自定义图形。

更多
• ### 无纸记录仪

无纸记录仪将数据采集和显示站完全集成，具有安全的内置数据存储和网络连接，其解决方案将数据采集设备集成到控制面板中，并符合NEMA标准。

更多