Managing Sterilization through Calculation of the Fo Value

### Overview

 Sterilization heating temperature measurement

In the sterilization process during pharmaceutical and food manufacturing, the "Fo" value is used as a benchmark for the sterilization time. The Fo value expresses the heating time required for disinfection, and varies according to fluctuations in heat applied to the substance being sterilized. The DX automatically calculates this Fo value from the heating temperature (the temperature measured on the DX), and is useful in managing the sterilization process by displaying and recording the value together with the heating temperature. Such careful monitoring and control allows minimization of sterilization time while maintaining the nutritional properties of the food.

Customer Needs

• To automatically calculate the Fo value, record it along with heating conditions, and use these as QC data
• To monitor the Fo value by heating temperature, shorten the heating process, and stabilize food quality
• To install a simple, low cost, highly reliable system without using external instruments or computers

### Process Outline

 Displays and records the heating temperature and Fo value

The heating temperature of the substance being sterilized is measured on the DX Paperless Recorder. The Fo value is automatically calculated according to the heating temperature (temperature measured on the DX) by the DX's mathematical functions. Computed results are displayed and recorded together with heating temperature and other parameters, and saved as quality data.

Also, the DX/AS1 supports 21CFR Part 11, and allows you to save formal pharmaceutical sterilization management data for approvals.

Definition of the Fo Value
Fo= Δt ∑ 10(T-Tb)/Z

Δt Measurement interval Heating temperature (temperature measured on the DX) 121.1°C (for steam pasteurization) Temperature unit of logarithmic sterilization capability changes (generally, 10°C is used).

Example of Calculation of the Fo Value on the DX
31= (K01*(K03**((01-K02)/K03)))/K04+31

01 Heating temperature (temperature measured on the DX) Fo value calculation result 1 sec or 0.125 sec (DX's measurement interval) 121.1°C (Fixed. Equivalent to Tb in the Fo equation) 10°C (Fixed. Equivalent to Z in the Fo equation) 60 (sec, for conversion to units of minutes)

### Yokogawa's Solution

 DX

DX series paperless recorders capable of automatically calculating the Fo value.

• Fo value calculated automatically by a math function (optional)
• Measured and calculated results saved electronically to local media while being displayed in real time on the large, color TFT display
• Also enables data sharing on the network (Ethernet compatible)

### Conclusion

DX series instruments allow you to automatically calculate and process Fo values as manufacturing/quality data at a reasonable cost. Also, by constantly ascertaining the Fo value according to the heating temperature, you can optimize (shorten) the heating process, and in the case of foodstuffs, minimize excessive heating which can damage the nutritional elements in foods.

• ### 制药

能否充分利用大型新兴市场的机遇，是目前制药行业面临的重大挑战。现在制药公司比以往任何时候都更需要引入精益化生产技术，以提高盈利能力。作为全球知名工业自动化供应商，横河电机致力于为客户的优化生产实践提供卓越的解决方案。

更多
• ### 食品和饮料

在食品生产企业，除了应对“确保食品安全和安全”、“降低成本”这些传统挑战，生产厂商还需要应对病菌和减少劳动力。
横河公司通过密切合作、共同应对挑战、提出适合运营的理想工厂，帮助生产厂商创造舒适的工作环境。

更多

• ### 底盘可拆卸型DX1000N

DX1000N特点是仪器的前面板有一个可拆卸的底盘，这使得DX1000N的内部组件易于连结和维护。

更多
• ### 按钮操作式DX1000/DX2000

Daqstation DX1000/2000系列是带显示的数据采集系统。HMI支持工业应用、审计跟踪和符合FDA 21CFR Part11高级安全性的自定义图形。

更多
• ### 无纸记录仪

无纸记录仪将数据采集和显示站完全集成，具有安全的内置数据存储和网络连接，其解决方案将数据采集设备集成到控制面板中，并符合NEMA标准。

更多
×