KOYAMA Etsutaro1 NIKKUNI Masaaki1 ODOHIRA Tetsu1
For a long time now, we have appreciated the standard 4-20 mA signal for 1to-1 analog communication lines in various plants. The time has now come for a new type of communication, namely, fieldbus, which features digital transmission on 1-to-1 communication lines. We have just developed an EJA series differential pressure transmitter with FOUNDATION™ fieldbus communication to be used as one of the main sensors in a plant. The fieldbus communication is to be supported by all models of the EJA series. That is, all new EJA models are equipped with the Standard Resource Block and the Standard Analog Input Block.
This paper describes the structure and the characteristics of the EJA transmitter with fieldbus communication.
- Industrial Measurement & Control Business Division
|Figure 1 External View
of the EJA Series
Differential pressure transmitters, which serve as the core of field instrumentation, are widely used to measure the flow rate, pressure and density of liquids, gases and steam, as well as the level of liquid in a tank.
There is strong demand from industrial plants for more accurate plant operation and labor savings in both plant operation and management as a means of improving yields. Not only are field devices required to be more precise and stable, but they must also be more intelligent in order to cut the cost of labor for plant operation and maintenance. In 1988, Yokogawa Electric released their first pressure transmitter with the BRAIN communication function-a communication protocol in which the 4-20 mA analog signal is mixed with digital signals. Since then, they have continued to release newer models of these pressure transmitters as well as striving to produce increasingly intelligent pressure transmitters, such as the DPharp series.
Recently, the focus has turned to the standard fieldbus (with its multidrop connection and full digital transfer) that will replace the conventional BRAIN communication with 1-to-1 connection. This paper introduces the EJA series pressure transmitter that supports the fieldbus protocol advocated by the Fieldbus Foundation. Figure 1 is an external view of the transmitter.
DESIGN CONCEPTS AND FEATURES
We now describe the design concepts and features of the fieldbus-enabled EJA series pressure transmitter.
Figure 2 Components of the Transmitter
The EJA series transmitter has three blocks for connecting itself to other devices: one resource block and two analog input blocks. These function blocks are configured using the standard parameters specified by the Fieldbus Foundation. Consequently, users can use the EJA series transmitter without needing device information unique to the transmitter. We believe that the ease at which the transmitter can be connected to the fieldbus is its key strength. It also goes without saying that the specific functions of the EJA series transmitter are also possible if that transmitter's own device information (Device Description, or DD) is used.
- High Precision, Stability and Reliability
As an EJA series device, the fieldbus-enabled transmitter model inherits the existing industry-proven features of that series, namely, the high-precision, high-stability and high-reliability characteristics.
All key components of the transmitter are Yokogawa originals that have been fabricated under strict quality control. The fabrication process involved both automation and mechanization in order to manufacture products with less product-to-product variation in performance.
The conventional analog transfer technology is designed to output only one signal for the control information. Fieldbus technology, on the other hand, permits the use of multiple information items such as status information, alarm information, the PV (primary value) and the SV (secondary value).
The EJA series transmitter outputs the PV value (the value for differential pressure) and SV value (the value for static pressure), along with the status information. It can also transmit alarm signals if the process value exceeds the limits (setpoints) for plant operation. Furthermore, the transmitter can send out alarm signals once the operating ranges defined in the transmitter capsule specifications are exceeded or if it detects any failure as a result of the sensor's internal judgment or the memory status.
- Compatibility with the Existing Capsules
The new model is compatible with existing EJA series transmitters, which means that a conventional model based on the BRAIN (4-20 mA analog transfer) protocol can be changed to a fieldbus-enabled model by simply replacing its converter.
- Reduction in Instrumentation Costs
Bus connection enables users to reduce the cost of instrumentation even below that of the conventional 1-to-1 connection. Furthermore, an additional transmitter can be easily installed next to an existing fieldbus-enabled device at any time in the future by connecting an additional cable from the device to the transmitter.
Figure 2 shows the components of the differential pressure transmitter. The transmitter can be roughly divided into the capsule and the transmission block. The capsule comprises the pressure sensing unit, cover flange and process connector. The transmission block comprises the converter and terminal box and can be commonly used for all models in the EJA series. In addition, the transmitter can contain an optional LCD display for on-site monitoring.
1. Circuit Configuration
Figure 3 is a circuit diagram of a fieldbus-enabled EJA series transmitter.
Figure 3 Circuit Configuration
The circuit is divided into two blocks: an excitation circuit block that excites the resonant sensor to detect its eigenfrequency and a converter block that applies a computational process to signals in order to convert them into signals for output to the fieldbus. The converter block measures the signal frequency using a dedicated microprocessor and the time interval using a reference clock, so as to improve the resolution. This measurement is carried out continuously to further improve the resolution by means of averaging. Corrective computing is carried out so that a precise reading of differential pressure can be obtained.
The transmission block incorporates Yokogawa's proprietary MAU device (ASIC for fieldbus applications) and a communication controller for implementing fieldbus communication.
Figure 4 Functional Block Diagram
Figure 4 shows a functional block diagram of the transmitter. The functions of the fieldbus-enabled EJA transmitter model are classified into the following blocks:
- Resource block
- Transducer block
- Analog input block
Each block is described below.
- Resource Block
This block manages the common resources for shared use with each function block. For example, an alarm for a failure in the non-volatile memory comes from this block.
- Transducer Block
This block calculates the current differential pressure and static pressure values to transfer them to the analog input. It detects the frequency signal from the sensor and carries out corrective computations using the temperature and static pressure signals. In addition, the block adjusts positioning errors using tuning parameters in order to calculate the differential pressure.
The resource and transducer blocks of the EJA series transmitter are designed to comply with the standard specifications. Thus, all of the functions unique to the transmitter are incorporated in this block. For example, the selection of data for the LCD display is carried out in this block, as is the judgment for alarms for out-of-limits operation and device failures.
- Analog Input Block
This block receives the differential pressure (instantaneous pressure) calculated by the transducer block and calculates the transmitter's output value. More specifically, the block carries out pressure range scaling, square root extraction (for flow rate calculations), output-scale calculation, filter calculation, and so on.
The calculated output is used as the transmitter's output and transferred to other field devices along with status information. The EJA series transmitter has two analog input blocks, each of which are designed to separately perform the computations noted above for both differential and static pressures.
In this paper, we discussed the features, structure and functionality of the fieldbus-enabled transmitter of the EJA series. This recently developed model is provided with the common functions of the fieldbus standard and prioritizes the need for an easy transition from the conventional instrumentation method. We are confident that this product will help spread fieldbus technology and enable future plants to achieve labor savings and precision operation.
As technical infrastructures such as engineering tools improve and fieldbus technology spreads, we believe that more advanced functions, which require only simple work operations, will be increasingly possible. We continue to monitor market trends and strive to offer solutions that meet an even wider variety of user demands.
- DPharp is a registered trademark of Yokogawa Electric Corporation.
- Saegusa T., et al., "DPharp Electronic Differential Pressure Transmitters," Yokogawa Technical Report, Vol. 36, No. 1, pp. 21-28 (1992).
Floating Production Storage & Offloading (FPSO)
- Floating Production, Storage, and Offloading (FPSO) unit
- Offshore production facility
- Stores processing equipment, produced hydrocarbons storage
- Feasible to produce oil in remote areas, deeper water
Oil & Gas Midstream
Yokogawa's innovative technology platforms and industry leading execution is well regarded in the global marketplace as a solutions partner pioneering the integration of technologies for all aspects of the oil & gas ecosystem from reservoir to enterprise. Proven solutions that include predicative business modeling, plant optimization and highly reliable automation platforms are assisting midstream operators to run their businesses at the optimal efficiencies. Yokogawa is helping clients to develop their automation roadmaps ensuring years of highly effective asset utilization and sustainability.
The onshore upstream industry faces ever higher demands and greater challenges with increasingly difficult and hostile environments in which it must operate.
In the mid 1970s, Yokogawa entered the power business with the release of the EBS Electric Control System. Since then, Yokogawa has steadfastly continued with the development of our technologies and capabilities for providing the best services and solutions to our customers worldwide.
Yokogawa has operated the global power solutions network to play a more active role in the dynamic global power market. This has allowed closer teamwork within Yokogawa, bringing together our global resources and industry know-how. Yokogawa's power industry experts work together to bring each customer the solution that best suits their sophisticated requirements.
Processing & Fractionation
- Natural-gas processing
- Control dew point of natural gas stream
- Separate natural gas liquids for sale and distribution
- Topsides processing and handling on the production platform prep extracted hydrocarbons for transport
- Yokogawa integrated control/monitoring maximize topsides operations productivity and availability
Water & Wastewater
Water resources are finite, and therefore contributing to a sustainable water cycle is one of the Sustainable Development Goals (SDGs). Yokogawa has been providing advanced digital control solutions for the stable supply of clean and safe water, wastewater treatment for protecting the water environment, water loss management, and optimization of plant operation for reducing CO2 emissions and running costs. Yokogawa works with you to increase your efficiency while ensuring compliance standards …. Smart measuring IA to IA.
Wellhead & Separation
The wellhead provides the structural and pressure-containing interface for the drilling and production equipment. The surface pressure control is provided by a christmas tree, which is installed on top of the wellhead. Isolation valves and choke equipment physically control the flow of well fluids when the well goes into production. Various automation packages are added to locally or remotely monitor, control and optimize production from each well or multi-well pad. Oilfield separators employ a pressure vessel to separate well fluids produced from oil and gas wells into gaseous and liquid components which are then transferred to pipelines or localized storage depending on the areas infrastructure.
Related Products & Solutions
Traditional-mount Differential Pressure Transmitter based on the EJA-A Series.
Differential Pressure Transmitter attached to an IFO assembly based on the EJA-A Series.
Differential Pressure Transmitter with Remote Diaphragm Seals based on the EJA-A Series.
EJA210A / EJA220A
Flanged-mounted Differential Pressure Transmitter designed for Liquid-level applications based on the EJA-A Series.
Traditional-mount Absolute Pressure Transmitter based on the EJA-A Series.
Traditional-mount Gauge Pressure Transmitter based on the EJA-A Series.
Gauge Pressure Transmitter with Remote Diaphragm Seal based on the EJA-A Series.
Traditional-mount High Gauge Pressure Transmitter based on the EJA-A Series.
In-Line Mount Gauge Pressure Transmitter based on the EJA-A Series.