F-value Computation for the Sterilization Process for Pharmaceutical and Chemical Products

Overview

When sterilizing preserved foods by heating, bacteria must be completely annihilated without changing the heat of the product (a degradation of the taste) and the integrated fatality (F value) must be recorded and monitored. By using the computation function of the SMARTDAC+ series Paperless recorder GX/GP (relational computation, four arithmetical operations, and exponential computation), the F value can be calculated and recorded.

• The F value at a specified sterilization temperature is automatically computed and recorded as production data.
• Using external input contact, recording can be started or stopped automatically.
• Automatic data file transfer and WEB monitoring are available.

F value

“F value” is The integrated fatality of the bacteria obtained by heat sterilization.  It is the integrated value of the fatality rate (Li) of the bacteria per unit amount of time (Δt),  and Li is a function of the sterilization temperature.

Configuration Example on the SMARTDAC+

Ex. Sterilization temp. Tr=121.1degC, Z value Z＝10degC,
Measurement interval Δt=0.0167min(=1sec)

• Constant
K1=10, K02=10.0degC(Z value), K03=121.1degC(Sterilization temp.), K04=0.0167min(measurement Interval), K05=100degC(measurement interval)
• Constant K2 to K5 may vary depending on the conditions.
• Equations
Measurement channel 0001
MATH channel A001：　K01**((0001-K03)/K02) ←Li value
MATH channel A002：（A002+A001*K04)*(0001.GT.K05) ←F value

*When the temperature falls to 100℃, the value of A002 is reset to 0.

Temperature monitoring for HACCP

In HACCP, the temp. management is required for the whole food supply chain (ex. Food manufacturing, processing, cooking, and sales). SMARTDAC+ series can record multiple points of temp. in production lines, facilities or environment of storage …etc.
It provides strong supports for your temp. management.

• Food & Beverage

In the food manufacturing industry, in addition to the conventional challenges of achieving both "ensuring food security and safety" and "cost reduction," ｍanufacturers are also working to cope with infectious diseases and labor saving.
YOKOGAWA supports ｍanufacturers to create comfortable workplaces by working closely, coping with challenges together, and proposing ideal factories suitable for operations.

Mehr anzeigen

• DX1000/DX2000 mit Tastenbedienung

Die Daqstation-DX1000/2000-Serie ist ein Datenerfassungssystem mit Anzeige. Sein HMI unterstützt benutzerdefinierte Grafiken für Industrieanwendungen, Audit Trails und erweiterte Sicherheit nach FDA CFR21 Part 11.

Mehr anzeigen
• Papierloser Touchscreen Datenrecorder GX10/GX20

Der SMARTDAC+TM GX10/GX20 ist ein schalttafel-montierter papierloser Recorder mit einem modularen Aufbau auf der Rückseite und einer Datenerfassungsfunktion zu Erfassung der erforderlichen Daten. Da das Gerät nicht nur I/O, sondern auch viele andere Kommunikationsprotokolle unterstützt, können Sie es mit verschiedenen Geräten verbinden. Die KI-Funktion ist serienmäßig voreingestellt. Erfüllt FDA 21 CFR Part11 und AMS2750E/NADCAP.

Mehr anzeigen
• µR10000 & µR20000

Die μR-Serie der kompakten industriellen Papierschreiber mit Aufzeichnungsbreiten von 100 mm (μR10000) und 180 mm (μR20000).

Mehr anzeigen

Top