ITOU Akio1 MIMURA Shin-ichi1 KOYAMA Etsutarou1 ODOHIRA Tetsu1 NIKKUNI Masaaki1 MIYAUCHI Tatsuhiko2
We have developed the EJX910 multivariable transmitter, an all-in-one instrument that integrates the functions of a differential pressure transmitter, a pressure gauge, a thermometer and a flow computer, while featuring high performance and space-saving design. The transmitter employs a unique flow rate calculation method, achieving a mass flow calculation cycle of 100 milliseconds. By adopting a Reynolds number compensation algorithm, etc., all flow calculation parameters were optimized and a mass flow accuracy rate as high as 1% was achieved. Furthermore, EJX910 complies with a wide range of primary devices, including orifices, nozzles and venturi tubes, and various types of fluid, including general fluids, steam tables, and natural gas. Application information, such as the primary devices and fluid data required for mass flow calculation, is input using the EJXMVTool, a mass flow parameter configuration tool that runs on a PC and is downloaded to the transmitter by means of field communication. A field test performed on a British natural gas test line showed an excellent mass flow measurement accuracy of 1%.
- Industrial Automation Products Business Division
- Global Service Headquarters
|Figure 1 External View
When evaluating mass flow rates using a primary device such as an orifice or nozzle in a differential pressure flow meter in order to make fluid density compensations, the upstream pressure (static pressure) and fluid temperature are measured in addition to the output of a regular differential pressure transmitter. In the past in such cases, a differential pressure transmitter, a pressure transmitter, a temperature converter and a flow computer were all separately required. The DPharp EJX series of differential pressure transmitters developed by Yokogawa in 2004, can simultaneously measure both differential pressure and static pressure using an advanced form of silicon resonant sensors that comprise the multi-sensing function. Recently, we have developed the EJX910 multivariable transmitter as a new model that incorporates this series' functions to full advantage. The EJX910 integrates the functions of a differential pressure gauge, a pressure gauge, a thermometer, and a flow computer into a single instrument, thereby achieving high space efficiency and multifunctionality.
A Reynolds number compensation algorithm and other means have been adopted for the mass flow rate calculation of this transmitter to optimize all flow rate calculation parameters and achieve high-precision mass flow rate measurement. In this paper, we will focus on the functions related to mass flow rate calculation, one of the features a multivariable transmitter has to offer. Figure 1 shows an external view of the EJX910.
The EJX910 multivariable transmitter serves as a differential pressure gauge, a pressure gauge, and a thermometer (with an external temperature sensor). In addition to this multifunctionality, the fluid density compensation function provided by the transmitter itself and the PC-installed EJXMVTool's mass flow parameter configuration tool enable high-speed, high-precision mass flow rate measurement. The EJX910 supports a number of flow rate standards and a variety of fluid types as target applications. Moreover, the EJX910 can be applied to integrated flow rate measurement and various diagnoses that use many process variables (differential pressure, static pressure, temperature, etc.).
Mass Flow Rate Calculation
|Figure 2 Example of Mass Flow Rate
Measurement Using Orifice
|Figure 3 Mass Flow Rate Measurement
Figure 2 shows an example of measuring mass flow rates from orifices and temperature sensors installed in a process. The EJX910 measures the difference between the upstream and downstream pressures of the orifice placed in the process, the upstream static pressure and the fluid temperature. Then the transmitter calculates the mass flow rate from these measured values.
Figure 3 shows a configuration of a mass flow rate measurement system. Application information necessary for flow rate calculations (primary device and fluid information) is input using the EJXMVTool mass flow parameter configuration tool running on a PC. This information is then converted into parameters that can be perceived by the transmitter and downloaded to the transmitter by means of field communication.
Figure 4 shows a block diagram of a mass flow rate measurement system comprising the EJX910 and EJXMVTool. Differential pressure, static pressure and temperature measured by the EJX910 can be directly output as process variables. The system performs fluid density compensation calculations according to the following equation to determine the mass flow rate.
|Figure 4 Mass Flow Rate Measurement Block Diagram|
Qm: Mass flow rate
C: Discharge coefficient
β: Beta ratio (d/D)
d: Bore of primary device
D: Pipe inner diameter
ε: Gas expansion factor
∆P: Differential pressure
ρ1: Fluid density
For this purpose the system employs a unique method of calculation that minimizes the transmitter's calculation load, and achieves a flow rate calculation cycle of 100 milliseconds. In a simplified method of flow rate calculation, the parameters in Equation (1) are treated as fixed values, resulting in large mass flow rate calculation errors as shown in Figure 5. The EJX910 performs optimized calculations in real time using dynamically changing parameters, thereby realizing a high flow rate accuracy of 1%. More specifically, Reynolds number corrections are made to the discharge coefficient (C) according to changing measured values. The gas expansion correction factor (ε) is corrected against the effects of adiabatic expansion of gases. In addition, the fluid density (ρ1) is corrected for static pressure and temperature variations.
|Figure 5 Comparison of Mass Flow Rate Errors between
Fixed-parameter Calculation and Optimized Calculation
The primary devices that the multivariable transmitter is compatible with, i.e., orifices, nozzles and venturi tubes, comply with a number of flow rate standards. In addition, a fixed mode is available to set desired values to the discharge coefficient and the gas expansion correction factor. The transmitter can handle a wide variety of fluid types as described below:
- General fluids (gases and liquids)
The transmitter supports 12 types of general fluids and employs the DIPPR® physical property database widely used around the world.
- Steam tables
The transmitter supports IAPWS-IF97 Water and Steam (1997), the standard for steam tables used in each country (e.g., JSME and ASME).
- Natural gases
The transmitter complies with the following typical natural gas calculation standards:
AGA8 Detail Characterization Method
Gross Characterization Method 1
Gross Characterization Method 2
ISO 12213: 1997
First edition (Dec. 1, 1997)
Part 2: molar-composition analysis
Part 3: physical properties
- Custom fluids
User-maintained physical property values can be incorporated in the transmitter.
Structure of EJXMVTool Mass Flow Parameter Configuration Tool
|Figure 6 Structure of EJXMVTool Software|
Figure 6 shows the structure of the EJXMVTool mass flow parameter configuration tool used to set mass flow rate application information.
- Flow rate parameter configuration function
This function comprises interactive graphic screens used to configure the flow rate calculation parameters shown in Equation (1), including application information such as primary devices and fluid types. Parameters can be configured easily as this function automatically converts the parameters to the transmitter's format.
- Flow rate parameter management function
This function saves flow rate parameters to a file, uploads and downloads these files to and from the transmitter, obtains results of parameter calculation by the transmitter, and outputs reports.
- General-purpose parameter management function
This function sets general-purpose parameters such as ranges and units.
- Physical property database
The software contains a database of physical properties necessary to calculate fluid density and viscosity.
- Field communications server
For easy compliance with various methods of field communication, the software incorporates the same field communications server used in the PRM (Plant Resource Manager) field device management package.
Flow of Application Information Setting
|Figure 7 Settings of Primary Device Information||Figure 8 Settings of Fluid Density and Viscosity|
As explained below, the operating screens of EJXMVTool have been designed in association with specific applications.
- Screen for setting primary device information
As shown in Figure 7, this screen is used to set the diameter of an orifice or other primary device, the pipe diameter and the materials.
Figure 9 Verification of Results of
Flow Rate Parameter Calculations
This screen is used to select the fluid type to be measured.
- Screen for setting natural gas composition information
This screen is used to set the composition information of a natural gas.
- Screen for specifying fluid pressure and temperature range
This screen is used to specify fluid pressure and temperature range, which are necessary to calculate density or other data items.
- Screen for setting fluid density and viscosity
As shown in Figure 8, this screen is used to verify data to set fluid density and viscosity. This screen can be customized.
- Screen for downloading application information
This screen is used to convert user-input primary device information and fluid information into a transmitter-specific format and then download that information to the transmitter.
- Screen for simulated flow rate calculations
This screen is used to verify the results of simulating mass flow calculations by the transmitter with the sensor inputs in Figure 9 applied as simulated inputs, in order to predetermine the results of applying the downloaded information.
- Screen for verifying flow rate calculations in an actual plant
This screen is used to verify the results of flow rate calculation under actual plant operation with the respective sensor inputs in Figure 9 actually applied.
Field Test Results
Figure 10 Results of an Actual Natural Gas Flow Test
Figure 10 shows the results of an actual flow test performed on a British natural gas test line. The test results show a mass flow rate measurement accuracy level as excellent as 1%. Users have also highly evaluated the EJX910 in other actual flow tests.
In future we intend to further accumulate experience in mass flow rate measurement using multivariable transmitters, while broadening the range of applications to encompass an even greater variety of primary devices and fluids. In addition to mass flow rate measurement, we will further develop fluid level measurement and multivariable measurement-based process diagnosis.
- ISHIKAWA Tamaki, et al., "New DPharp EJX Series of Pressure and Differential Pressure Transmitters," Yokogawa Technical Report, No. 37, 2004, pp. 9-14
- KUROMORI Ken-ichi, "Industrial Flow Meters—Recent Topics—." Measurement & Control, Vol. 42, No. 12, 2003, pp. 1015-1020 (in Japanese)
- 'DPharp', 'EJXMVTool' and 'PRM' are registered trademarks of Yokogawa Electric Corporation. All other product names, including software names, appearing in this document are the trademarks or registered trademarks of their respective companies or groups.
석유 화학 제품, 무기물 또는 중간체를 생산하든 관계없이 화학 회사는 안전하고 호환되는 작업을 유지하면서 적시에 효율적인 방법으로 제품을 제공하는 비용과 마진 압박에 시달리고 있습니다. 또한 화학 회사들은 공급 원료 및 에너지 가격의 변동에 적응하고 가장 수익성 높은 제품 혼합을 시장에 제공해야 합니다.
Yokogawa는 벌크 화학 시장의 자동화 요구 사항을 세계적으로 지원해 왔으며 이 시장에서 인정받는 선두 주자입니다. Yokogawa는 제품, 솔루션 및 업계 전문 기술을 통해 시장 및 생산 요구 사항을 이해하고 플랜트의 수명주기를 통해 안정적이고 비용 효율적인 솔루션을 제공하기 위해 협력합니다.
1970년대 중반, Yokogawa는 EBS 전기 제어 시스템 (EBS Electric Control System)의 출시와 함께 전력 사업에 진출했습니다. 그 이후로 Yokogawa는 전 세계 고객에게 최상의 서비스와 솔루션을 제공하기 위한 기술과 역량의 개발을 꾸준히 지속해 왔습니다.
Yokogawa는 역동적인 글로벌 전력 시장에서 더욱 적극적인 역할을 수행하기 위해 글로벌 전력 솔루션 네트워크를 운영했습니다. 이로 인해 Yokogawa 내에서 보다 긴밀한 팀워크가 가능해져서 글로벌 리소스와 업계 노하우를 하나로 모았습니다. Yokogawa의 전력 산업 전문가들은 각 고객에게 정교한 요구 사항에 가장 적합한 솔루션을 제공하기 위해 협력합니다.
정수 및 하수 처리
Yokogawa는 보다 효율적인 사회를 만들기 위하여 에너지 최적화 기술 개발, 온실가스 감축, 오염 물질로부터 환경을 보호하는 견고한 제품 제작 등을 통해 지속 가능한 수자원 생산을 위한 제어 솔루션을 제공하고 있습니다. Yokogawa의 첨단 기술과 폭넓은 애플리케이션 노하우를 통해 고객과 협력하여 사업을 활성화하고 플랜트 수명주기 전반에 걸쳐 높은 가치를 제공할 수 있는 지속 가능한 솔루션을 제공합니다. 당사의 기술과 제품은 플랜트의 성능을 향상시키고 오늘날의 수자원 시장에서 경쟁적으로 운영할 수 있도록 보장하며 운영비를 절감합니다. Yokogawa는 지자체 및 공업용수 처리 시장에서 광범위한 수처리 응용 분야를 지원합니다.