Differential pressure (DP) flow measurement is a well-known and common technology for measuring flow in a closed pipe. DP flow solutions are made up of two elements to measure flow, a primary element, which introduces a pressure drop, and a secondary element (a DP flow transmitter) to measure the differential pressure flow. This technology uses Bernoulli’s equation to calculate the flow of the medium in the pipe.
Yokogawa offers a complete solution for measuring DP flow with a range of technologies, including averaging pitot tubes (VERIS Verabar®) and nozzle type flow meters (VERIS Accelabar®).
Los transmisores de Presión Corriente son un tipo único de transmisor diferencial diseñados para medir las pequeñas diferencias en ambientes donde la presión estática es muy baja.
Volumetric flow rate is the volume of fluid which passes per unit of time. Common units of measure include litres/sec (l/s) or cubic feet per second (ft3/s).
Differential pressure flow meter uses a variation of Bernoulli’s principle to infer the volumetric flow rate in a pipe. For our application of the principle, it states that the velocity of the flow increases and the pressure decreases as the flow moves through a restriction in the pipe; therefore, the pressure is higher before the restriction than after. Knowing this differential pressure, the flow rate can be determined. The square root of the differential pressure across the restriction in the pipe is proportional to flow.
Mass flow rate is the amount of mass of a substance which passes per unit of time. Common units of measaure include Kg/s or lbs/min.
This type of measurement needs more detailed input than just differential pressure. It also needs temperature, static pressure of the line, the composition of the fluid, and more information of the primary element.
Primary Element: The Primary Element generates the differential pressure in the fluid. Primary Elements can include an averaging pitot tube, an integral flow orifice (IFO), or any number of DP producing devices.
Secondary Element: The Secondary Element is the pressure transmitter that is used to measure the DP. The secondary element may or may not include a flow computer with temperature and static pressure input.
Tertiary Element: The Tertiary Elements consist of flow computers, gas chromatographs, static pressure transmitters, or temperature transmitters.
¿En busca de información adicional sobre Yokogawa Iberia, tecnología y soluciones?