That is possible but not guaranteed. The 225 mm sensor body is longer, so it has more electrolyte. Often the reference electrode fails because the electrolyte is depleted as it diffuses through the reference junction into the process. With a higher volume of electrolye this process will take a longer time. This is not the only failure mode so you can still have a short lifetime due to another cause.
The cation differential pH and ORP sensors were designed for difficult applications where conventional sensors are ineffective. These include measurements such as brine solutions to applications as diverse as electrolysis processes and cheese manufacturing.
Liquid analyzers are used for monitoring process chemistry including water quality, providing process optimization and control.
pH and ORP meters, analyzers and transmitters are used for continuous process monitoring of pH and ORP to ensure water/product quality, monitor effluent discharge, batch neutralization, pulp stock, scrubbers, cooling towers, chemical, water/wastewater treatment and many other applications.
pH electrodes and sensors are the sensing portions of a pH measurement. Various installation options including retractable, flow thru, immersion, and direct insertion. Proper pH electrode/sensor selection is critical for optimal measurement results.
Process analyzers are used to determine the chemical composition or physical properties of substances involved in industrial processes.