That is possible but not guaranteed. The 225 mm sensor body is longer, so it has more electrolyte. Often the reference electrode fails because the electrolyte is depleted as it diffuses through the reference junction into the process. With a higher volume of electrolye this process will take a longer time. This is not the only failure mode so you can still have a short lifetime due to another cause.
The cation differential pH and ORP sensors were designed for difficult applications where conventional sensors are ineffective. These include measurements such as brine solutions to applications as diverse as electrolysis processes and cheese manufacturing.
- Liquid analyzers
- Monitoring process chemistry
- Water quality
- Process optimization and control
pH and ORP meters, analyzers and transmitters are used for continuous process monitoring of pH and ORP to ensure water/product quality, monitor effluent discharge, batch neutralization, pulp stock, scrubbers, cooling towers, chemical, water/wastewater treatment and many other applications.
- pH and ORP sensors
- Proper pH electrode/sensor selection critical for optimal measurement results
- Installation options include retractable, flow thru, immersion, and direct insertion