pH Buffer solutions are an indispensable tool for maintaining an accurate pH measurement. A pH measurement loop requires regular calibration of the electrodes to compensate for their aging and deterioration, therefore it is essential to have a reliable standard with which to do this calibration. This is the task of pH buffers. They are the reference points to which the measurement electrodes are calibrated. Buffer solutions are used as a means of keeping pH at a nearly constant value in a wide variety of chemical applications. Consequently, if the buffers are not accurate themselves, the calibration serves no useful purpose.
The fundamental concept of buffers is that it is a solution designed to maintain its rated value even with the intrusion of contaminants. Buffer solutions are one which resists large changes in pH following the addition of either an acid (H+) or a base (OH-). This is because the weak acid and its conjugate base can react with both the acid and base. A buffer solution is an aqueous solution consisting of a mixture of a weak acid and its conjugate base or a weak base and its conjugate acid, mixed in a 1:1 ratio (equilibrium). These are called N.I.S.T. Buff ers. National Institute of Standards and Technology (NIST), formerly known as the National Bureau of Standards (NBS); is a federal technology agency that develops and promotes measurement, standards, andtechnology. The level of contaminants a Buffer can withstand before becoming inaccurate has a great deal to do with how close to equilibrium the solution is in the first place. The pH of such a solution is calculated by the following formula:
pH = pKa+log(salt)/(Acid)
If the Salt/Acid ratio is 1, then we have:
pH = pKa, because log 1 = 0
N.I.S.T. lists the pKa values for buffers at 25ºC:
4.008- Potassium Hydrogen Phtalate
6.865- Potassium Dihydrogen Phosphate/ Disodium Phosphate
9.180- Sodium Teraborate Decahydrate (Borax)
Therefore, the pH values for the most stable buffer solutions are 4.01, 6.865 and 9.18 respectively after rounding at 25°C. These are the equilibrium values at which the buffer can accept the maximum level of contaminants before becoming inaccurate. This is commonly referred to as the "buffer capacity." The constituents of these liquids are defined by international standards like DIN19266, IEC 726 and NIST.
What Buffer values do you use... 4.00, 7.00, and 10.00? If you use 4.00, 7.00, and 10.00 you are not using N.I.S.T. Buffers, you are using what are known as Technical or "Traceable to N.I.S.T." Buffers. You can tell a N.I.S.T. Buffer from a Traceable Buffer because no N.I.S.T. Buffer at reference temperature (25ºC) is a whole number and all traceable buffers are whole numbers (4.00, 7.00, and 10.00).
A Traceable or Technical Buffer is made by taking the N.I.S.T. formulations for true buffered solutions and adjusting them away from the 1:1 ratio to get the whole number, 4.00, 7.00, or 10.00. This was done 15-20 year ago because with the past old Analog Transmitter it was simpler and straightforward to read a whole number during calibration. Which would you be able to see better on the transmitter below in Figure 1, a pH of 6.86 or a pH of 7.00? The answer is simple 7.00 with out a doubt.
Examples of the N.I.S.T. buffers used by YOKOGAWA are shown in Table 1. Buffer solutions prepared from these substances conform to the recommendations of the DIN Standards Committee and the National Instit ute of Standards and Technology (NIST). The substances were chosen for their particular suitability as calibration standards for precision pH meters. Examples of the Traceable buffers are shown in Table 2. All Yokogawa pH Transmitters are preprogrammed with N.I.S.T. Buffer tables for calibration. Refer to Technical Note TNA0905 "PH450 Analyzer Programming Customer Buffer Tables", before calibrating a Yokogawa EXA PH series transmitter with Traceable or Technical Buffers; you will need to program th e buffer table to reflect the buffer table of the buffer you are using. Refer to Technical Note TNA0919 "PH450 Analyzer Free Programmable Buffer Tables" for help.
|EXA Auto CAL NBS Buffer Table|
|Typical NBS Traceable Buffer Table|
The temperature dependence of the pH of a buffer solution is generally specified in terms of measured pH values at certain di screte temperatures. Since the Yokogawa EXA PH series instruments are pre-programmed with N.I.S.T. Buffer tables, you can see from table 1 above that as the temperature ch anges the pH of the buffer solution also changes. So if during calibration the temperature compensator needs to be immersed in the buffer liquid, an automatic adjustment for temperature variations will be done in the transmitter. Any pH value listed is only meaningful if the measuring temperature is also specified. High caustic Buffers (9.18 and 10.00) are the most temperature sensitive buffer that can be used. pH 4 Buffer is th e most temperature stable buffer that can be used.
The SC25V is a pH sensor in a 12 mm design that includes an integral temperature element and a Liquid earth electrode.
Designed for 2-wire (loop powered) system configuration. Standard mA (4-20), HART, Profibus, and Foundation Fieldbus communication available.
The PH450 pH and ORP analyzer is a multivariable analyzer that combines pH with temperature and ORP (Redox) measurement, which can be utilized through different output functions: two mA current outputs, four independent SPDT contact outputs and HART.
The PH20, FU20 and FU24, all-in-one pH and ORP, sensors show how Yokogawa applies the motto "Simple is best" to sensor technology.
Yokogawa has invested considerable design and development time in producing a full range of fittings with particular emphasis on designs that reduce installation and maintenance time and consequently save operation costs.
On-line measurements often present extra challenges, especially when routine maintenance is required. The PR10 is ideally suitable for applications where the sensors must be removed without interrupting or shutting down the process. Without any special tools the PR10 can be retracted safely from the process at pressures up to 5 bar (72 psi). Using the PR10 allows us to place any dissolved oxygen sensor that has a PG13.5 connenction into a retractable assembly.
The immersion fittings are designed for either pH or ORP (Redox) measurements in tanks, open vessels and drains. They have a "hoisting cable" for easy maintenance.
The heart of a pH measuring loop is the electrode system. Yokogawa has designed a wide range of electrodes to ensure this heart keeps beating under the most severe conditions.
Many botech and pharmaceutical applications for determining the purity and quality of the finished product, require the measurement of pH and conductivity in very low volume samples. The PC Flow Block assemblies were specifically designed to achieve accurate pH and conductivity measurements in such low volume applications.
The PH71/PH72 pH and/or ORP meter is a Compact, Easy-to-use, Drip proof – Ideal for Field Use. Features "One-Touch Calibration" and Temperature Compensation. Laboratory-Grade Intelligent pH Meters – Sized and Priced to Fit in Your Pocket
A pH electrode is as good as its maintenance; this is especially true in harsh environments. For industrial applications and particularly for automatic process measurements it is of the greatest importance that the sensitive part of a glass electrode and the diaphragm of a reference electrode are kept clean.
The PH87 retractable assembly is suitable for "Hot-Tap" service, allowing for removal from a pressurized process line or vessel without interrupting the flow or draining the tank. The housing assemblies are constructed of chemically resistant titanium sheathes and are used with Model PH97 rugged sensors.
These sensors feature the latest innovations in pH glass formulation, reference systems, and reference electrolytes. They offer supperior accuracy and stability, even after repeated sterilizations.